I am stuck in being able to calibrate the avisoft. I have a soundtrap Hf300 and I'm not sure how to insert the data into the avisoft calibration. Any advice? Thanks All the best
First, input the end-to-end calibration value for whichever mode you recorded your files in. Assuming you did your recordings in 'high gain' mode, then you'd put -176.4 as our "reference SPL" in Avisoft.
Then, check out page 19 of the soundtrap user guide. While it does not specify how Avisoft reads in data (and I am unfamiliar with Avisoft), it tells you how to import it into PAMGuard, Matlab, PAMGuide and Audacity, which could be helpful.
I would guess that you'd put 1 Vp as the drop-down selection for "full-scale range"... any Avisoft-experienced folks out there?